Keithley Instruments, Inc. warrants this product to be free from defects in material . aspects of Nanovoltmeter operation, refer to the Model User’s Manual. A. Nanovoltmeter. Accessories Supplied. Low Thermal Input Cable with spade lugs, m (4 ft). User manual, service manual, contact cleaner. View and Download Keithley user manual online. Nanovoltmeter. Measuring Instruments pdf manual download. Also for: a.
|Published (Last):||19 June 2009|
|PDF File Size:||12.58 Mb|
|ePub File Size:||2.26 Mb|
|Price:||Free* [*Free Regsitration Required]|
Keithley A Manuals
This procedure assumes that the case cover is already removed. Table of Contents Add to my manuals Add. Page of Go. To select a baud rate, follow these steps At 5 PLC, current can be reversed after msec. Low Thermal Input Cable with spade lugs, 1.
KEITHLEY SERVICE MANUAL Pdf Download.
Response 0 Calibration unlocked 1 Calibration locked Description command allows you to lock out calibration after completing: Lsync line Cycle Synchronization KEY is sent to the computer.
Keep in mind that the I-Source of the SourceMeter is a constant current source.
Test Circuit green CH 2 Ice Bath white Cable-to-thermocouple wire connection one of two Voltage and temperature connections Channel 1 should be used for voltage measurements since it supports a wider range of measurements, leaving Channel 2 to measure temperature. The Relative feature of the Model can be used to null out the offset voltage.
While displaying the Ratio result, disconnect the current source from the network. Valid values for Min is 1 toand valid values for Max is 2 to Additional information on these commands is provided after the table.
Buffer Recall Perform the following steps 212a view stored readings and buffer statistics: Note that every reading is automatically stored in the buffer. Doesn’t that kinda raise a flag?
Page 98 – Figure Delta measurement using bipol Adjust its position by gently pulling it away from the sides of the instrument case and swinging it up or down.
When I read out the buffer, the vi returns an array with all elements the same, and equal to the first element in the buffer wich I can read out from the keithley by using the front panel. In addition, recalled data includes statistical information minimum, maximum, peak-to-peak, average, and standard deviation. Testing Superconductor Materials Be sure to restore factory front panel defaults, and perform a full auto-calibration ACALas outlined 2182s.
Keithley 2182 Service Manual
The Model reacts to this mabual if it is the programmed control source. Line Frequency 50Hz or Hzwhile 60 indicates that it is set for 60Hz.
Turn off the power, and disconnect the line cord and all other test leads and cables from the instrument. Page 44 The thermocouple wires are connected to the copper lugs of the supplied input cable.
Remote Operation Condition registers Figure shows, some register sets have a condition register. Figure Measuring Switch Contact Resistance The Keithley A Nanovoltmeter is optimized for making stable, low noise voltage measurements and for characterizing low resistance materials and devices reliably and repeatably.
However, the device action block could include the following additional actions refer to Figure External scanning example with Model When Ratio is enabled, the state on or off of the REL annunciator depends on which measurement function was last selected.
Taking Readings Using The: INIT commands will not execute until calibration is completed.
Providing the finest test equipment solutions since 1992
Page 39 Getting Started To restore factory or user settings: Nulling thermal EMFs — To maximize handling speed, quick-disconnect test clips are typically used for the resistor connections.
Section 2 of this manual covers detailed calibration procedures. Overview The Analog Output voltage is calculated as follows: Query state of Front Autozero.